Find great deals for National Instruments NI USB Low-cost Multifunction DAQ. Shop with confidence on eBay!. The NI USB is a USB based data acquisition (DAQ) and control device with analog input and output and digital input and output. The price is NOK. : National Instruments USB Bit, 48kS/s Low-Cost multifunction DAQ: Computers & Accessories.
|Published (Last):||7 March 2007|
|PDF File Size:||17.64 Mb|
|ePub File Size:||3.45 Mb|
|Price:||Free* [*Free Regsitration Required]|
Get the item you ordered or get your money back. Digital input DI and digital output DO: Tasks and virtual channels can be given any name e.
TechTeach: A quick guide to National Instruments USB and USB multifunction I/O devices
Before you start using the USB, you should attach labels to the analog and digital terminals. Select a valid country. I am sure that one of them is the signal but which one? In this example the analog output channel AO0 and the analog input channel AI0 are connected, as in the loopback testing procedure described earlier.
Wire nuts may also be used.
This amount is subject to change until you make payment. Message 4 of Tested, fully functional, and in excellent physical condition.
Seller assumes all responsibility for this listing. Hopefully the self-test passes without errors.
National Instruments NI USB 6009 Low-cost Multifunction DAQ
To connect the four devices you will need to connect all of the brown wires together and connect to ground. Seller information optoscience The item may have some signs of cosmetic wear, but is fully operational and functions as intended. We may then repeat this procedure for other channels.
To prepare for the loopback test, AI0 and AO0 are wired together. Add to watch list Remove from watch list.
National Instruments USB Data Acquisition Card, NI DAQ, Multifunction | eBay
A local virtual channel exists only within it’s parent task, while a global virtual channel may be used in several tasks. I read your discussion. These 12 channels are organized in ports, with Port 0 having lines 0.
Assuming that MAX recognizes the device, you can now run a self-test of it by selecting 60099 in the menu. More free tutorials are available at http: In the following only the name USB is used. The test procedure, which is denoted loopbackis to connect the AI0 channel to the AO0 channel.
Window where you select the measurement type.
National Instruments USB data acquisition module. This shows the physical channel associated with the virtual channel, see Figure Excuse my lack of knowledge in this area. This seller offers shipping discounts on combined purchases for eligible items.
Contact the seller – opens in a new window or tab and request a shipping method to your location. In the Voltage Output dialog 60009select any voltage between 0V dwq 5V. Thanks for your prompt reply. See terms – opens in a new window or tab.
The item you’ve selected was not added to your cart. Learn more – opens in new window or tab Seller information optoscience Get an immediate offer.