D CV. Read this owner’s manual thoroughly before use Owner’s operation manual for series. DT Series. 3 1/2 Digital Multimeter. Read this owner’s manual thoroughly before use. DT series multimeter have been designed according to IEC concerning electronic measuring. View and Download Histest DT owner’s operation manual online. DT Multimeter pdf manual download.
|Published (Last):||4 December 2011|
|PDF File Size:||1.17 Mb|
|ePub File Size:||5.29 Mb|
|Price:||Free* [*Free Regsitration Required]|
Operating Instruction “COM” jack. If the defect has been caused by a misuse or abnormal operation conditions, the repair will be billed at a nominal cost.
Set the range switch to the desired range. Before attempting to open the case, always be sure that test leads have been disconnected from dt830c circuits. To use safely, each measurement can not last for more than manuzl seconds, and the interval between each measurement must be more than 15 minutes.
Disconnect circuit power and discharge all high-voltage capacitors before testing resistance, continuity, diodes, or capacitance. The voltage is approximate 5Vp-p. Front Panel Description This function allows the meter to output a signal as a signal generator.
Related Manuals for histest DT No related manuals. Comments to this Manuals Your Name.
In the latter case, the transient stresses are variable; for that reason, its necessary that the transient-withstand -capability of equipment is made known to the user. If the resistor to be measured is connected to a circuit, disconnect circuit power and discharge all capacitors before measuring resistance.
Page 6 To use safely, each measurement can not last for more than 10 seconds, and the interval between each measurement must be more than 15 minutes. Close case and tighten screws completely before using the meter to avoid electrical shock hazard. Page of 13 Go.
Histest DT830 Owner’s Operation Manual
Enter dh830d from picture: This function allows the meter to output a signal as a signal generator. A coupling capacitor should be used when connecting the meter to a circuit. Battery And Fuse Replacement Before attempting to open the case, always be sure that test leads have been disconnected from measurement circuits. Don’t have an account?
By pressing ‘print’ button you will print only current page. Don’t show me this message again.
Owner’s operation manual for DT Series 3 ½ Digital Multimeter
To print the manual completely, please, download it. Use the proper terminals, function, and range for your measurements.
Before measuring current, check the meter’s fuses and turn power off to the circuit before connecting the meter to the circuit. A test signal will appear between the and “COM” jacks. Set the range switch to ” ” or ” ” range.
Table of Contents Add to my manuals Add. Safety Information If the defect has been caused by a misuse or abnormal operation conditions, the repair will be billed at a nominal cost. Got it, continue to print.