The ELx continues our tradition of offering the tightest specifications of any microplate reader. Visit BioTek to learn more about our precision microplate. ELx Operator’s Manual. Revision History. Rev Date Changes. A. 1/96 First Release. B. 5/96 Revised reader specifications. Added Scanning. View and Download Biotek ELx operator’s manual online. Absorbance Microplate Reader. ELx Laboratory Equipment pdf manual download. Also for.
|Published (Last):||15 October 2008|
|PDF File Size:||7.61 Mb|
|ePub File Size:||7.69 Mb|
|Price:||Free* [*Free Regsitration Required]|
Lamp may be out or blocked by a faulty filter wheel.
Intended Use Statement Don’t have an account? This average is used to calculate the standard curve instead of using the individual replicate of each standard. Installation Page 32 – Product Registration Page 33 – 2: Use the formula keys Math, Other, Map and Function to define the.
Prompts For Well Location Don’t show me this message again. Replace the cover, and verify that the white plastic indicator peg lines up with the correct voltage indicator hole. Set ekx808 date and time.
Select The Operating Environment BioTek Representative or BioTek. Examples of mapping and replication directions are shown on the next page. Liquid Test 3 is an optional test offered for sites that must have proof of majual at wavelengths lower than those attainable with the Absorbance Test Plate. Because of the mathematical correction, this difference under normal conditions should be slight or nonexistent. Page 35 The wheel should easily disconnect from the hub.
Page 30 Chapter 1: The wheel is shipped in a separate envelope.
Biotek ELx808 Operator’s Manual: Error Codes (general)
Getting Started With Gen5 Perform a regression analysis on the data to eelx808 if there is adequate linearity. Check filter wheel for empty positions.
Errors and indicate that a motor was not able to move to its. Onset Od Selection To properly store interference filters for the ELx during extended periods of non-use, package the filters in a light-tight envelope or container, away from high humidity and direct sunlight. Column Report Without Samples ELx The following specifications apply only mqnual standard well, flat- or round-bottom microplates.
Usa Fcc Class A Restrictions And Liabilities Calculate Repeatability test B Install the Filter Wheel Page 35 Page 36 – 4: Perform Liquid Test 1 Product Support And Service Errors and indicate that a particular axis has moved to a point.
Biotek ELx808 Manuals
The Absorbance Plate Test tests the accuracy and repeatability of the reader to 2. Kinetic Number Of Reads Table Of Contents Some readers have custom programmed assays installed.
The top line of the display updates to reflect this selection.
Transformation Formulas Divide all ODs on the plate by 2 and multiply by Autocal error, see error Electromagnetic Interference And Susceptibility Failed configuration checksum test. The display updates manul reflect your selection.
Got it, continue to print. Reading A Microplate The software will automatically increment each subsequent sample identification number by 1.
BIOTEK ELX OPERATOR’S MANUAL Pdf Download.
Incubator Self Test Current Average: Only qualified technical personnel should perform service procedures on internal components. Defective motor drive circuit or volt power problem.
To define a cutoff formula for Positive and Negative calls: Recommended Maintenance Schedule Chapter 5: Page The values obtained for all the controls should be in the ranges specified on the QC Certificate. Got it, continue to print.