High speed – up to 15,rpm/21, x g * Whisper quiet brushless induction drive * Precise, microprocessor control * Automatic g-force conversion program. Hermle ZM-2 Microcentrifuge from Labnet International, Inc. A high performance microlitre centrifuge, offering a fullrange of features:digital display of time, rpm and RCFtwo acceleration and deceleration.
|Published (Last):||9 January 2011|
|PDF File Size:||6.71 Mb|
|ePub File Size:||2.50 Mb|
|Price:||Free* [*Free Regsitration Required]|
Service And Inspection Of Centrifuge 7. Preselection Of Operating Time 4.
Before operation, secure the rotor lid to the rotor by pressing the snap Before operation, secure the rotor lid to the rotor by pressing the snap Before operation, secure the rotor lid to the rotor by pressing the snap Before operation, secure the rotor lid to the rotor by pressing the snap Description – The rotor is not installed securely on the motor shaft.
There is hermlf distinction between two different kinds of errors.
During a run and with the centrifuge lid closed you have to press the “preset” key 1hold it and change operating time by turning the knob 3. The digital indications on the display will light up when the power is on. Don’t have an account? If this is not done, they could scratch the protective coating of the rotor. If the green “slow”control lamp is lit, the unit accelerates and decelerates very softly below 2 rpm.
Hermle LaborTechnik GmbH – Z 233 M-2 Microlitre Centrifuge
Comments to this Manuals Your Name. Got it, continue to print. Page 15 – Temperture Page 16 – Service and inspection of centrifuge Page 17 – Cleaning of centrifuge after breakage of Set Or Change Rotor-code -Load the rotor symmetrically by inserting tubes into spaces opposite from each other.
General Information The Hermle Z M-2 is neither explosion proof nor inert gas shielded and should, therefore, never be operated in explosive-hazardous locations. By pressing ‘print’ button you will print only current page.
Centrifuges Z M-2 -Hermle- Germany
Page 14 – Keyboard – Starting the centrifuge – “qu Overcurrent or undervoltage due to z23 supply fluctuations Action To open the centrifuge lid. Page 19 Error no.: Table of Contents Add to my manuals Add.
Figure 8 If the green “fast” control lamp is lit, the unit accelerates and decelerates fast. Operation Operation Operation Operation Operation 4.
G131038 Hermle Z 233 M-2 Labnet Centrifuge W/rotor
Should this happen, the centrifuge, rotor, buckets, adapters and the mm-2 chamber must be thoroughly cleaned and all broken particles removed immediately.
Commercially available tubes or bottles made of glass or plastic. To print the manual completely, please, download it. Never stay in the safety zone of 30 cm around the centrifuge or deposit dangerous materials inside this zone during centrifugation. The green control lamps 3 show which mode is activated.
Hermle z m-2 Manuals
The rotor must always be balanced in this fashion prior to operation. With the centrifuge lid open you can preset the operating time herme the knob 3.
Open and reclose the lid, restart the centrifuge Error no.: The Hermle Z M-2 is neither explosion proof nor inert gas shielded and should, therefore, never be operated in explosive-hazardous locations. Page of 20 Go. The temperature rise depends on the rotor swing-out or angle rotor ,bucket type, ambient temperature, running time and the speed of the rotor.
Page 20 Error no.: Use only original spare parts!
Power failure while rotor is in motion Action: Width 28 cm Depth 37 cm Height Keyboard – Starting The Centrifuge – “quick”-key 4. Enter text from picture: On the digital display “time min-sec” the letters “ro” rotor appears.
Turn the speed knob 1see figure 4 till the required rotor code appears on the d i s p l a y. Don’t bermle me this message again.