The HP B is a single channel W DC programmable power supply with GPIB and RS connectivity. It is part of the HP x series. Agilent B/B/B/B System DC Power. Supplies All references to HP-IB have been changed to GPIB. A note about. I just picked up two HP B power supplies as I read somewhere quickly that it not only can be programmed via GPIB to set voltage/current.
|Published (Last):||27 November 2008|
|PDF File Size:||1.27 Mb|
|ePub File Size:||18.9 Mb|
|Price:||Free* [*Free Regsitration Required]|
Bench OperationA fan cools the dc source by drawing air in through the sides and exhausting it out the back. Press Save, 1, Enter.
Agilent / HP B Watt DC System Power Supply 20v 5a | eBay
C – Error Messages—Invalid suffix [unrecognized units, or units not appropriate]—Suffix not allowed—Invalid character data [bad character, or unrecognized]—Character data too long—Character data not allowed—String data error—Invalid string data [e. Get fast answers from reviewers. HP Hewlett-Packard B user manual. System KeysThis is the blue, unlabeled key, which is also shown as Shift in this guide.
Anyhoo, the unit showed up all kinds of Dirt McGirt: The output of the dcsource may be adjusted to any value within the boundaries shown. In general, if the wires are heavy enough to carry the maximum short circuit currentwithout overheating, excessive voltage drops will not be a problem. One nice thing is that the current meter on it is fairly precise.
What’s missing? Tell us about it.
After much swearing, compressed air, and yp, the inside of the chassis looked like this:. Page 35 5Front panel OperationIntroductionHere is what you will find in this chapter: Release the key after you press it. Fast mode lets you improve or enhance certain operating characteristics; while at the sametime degrading other operating characteristics.
English Choose a language for shopping. If you want to quickly view the content of pages found on the following pages of the manual, you can use them. Page 3 Safety SummaryThe following general safety precautions must be observed during all phases of operation of this instrument. 6623b also made me less worried that there were secret vermin in my apartment that were taking refuge in my bench. CCThe output is operating in constant current mode. It still seems to be in cal from 5 years ago. Still, if your device is significantly out of cal, you can get a lot closer.
The dc source must be in set to Local mode to use the front panel controls. V or A MAX Although my original post about this is pretty old I still use this same unit as my main bench supply.
Not perfect, but not bad for an eBay bench supply! Set the instrument to the operating state that you 6632n to save. Page 26 3 – Installationat the load.
Refer to the Safety Summary page at the beginning of this guide for general safetyinformation. Disconnect all loads from the dc source. Please enter a question. For verification tests, see appendix B. Front Panel Operation – 52 – Querying and Clearing Output ProtectionThe dc source will disable its output if it detects an overvoltage or overcurrent fault condition.
Page 31 4Turn-On CheckoutIntroductionSuccessful h in this chapter provide a high degree of confidence that the dc source is operatingproperly. Selftest error messages appear as: Page 50 A – SpecificationsSupplemental CharacteristicsTable A-2 lists the supplemental characteristics, which are not warranted but are hpp oftypical performance determined either by design or type testing.
Disabling it will result in some disabled or missing features.
The fault FLT output,also referred to as the DFI discrete fault indicator signal in the front panel and SCPI commands,is an open collector circuit that pulls the positive output low with respect to the negative chassisreferenced common.
B – Verification and CalibrationTable B- 5. Tell us about it. Scroll through the Output menu by pressing q. Theoutput dropped to zero, CV turned off, and Protturned on. Function KeysImmediate Action KeysImmediate action keys immediately execute their corresponding function when pressed. Verification and Calibration – BTable B- 3. The display returns to Meter mode when the calculation is complete. If you read through the manual, you know, that it can also behave as current sink load.
The dc sourceis shipped with its HP-IB address set to 5. Or point us to the URL where the manual is located. Press Shift, OV,Enter8. Failure to comply with these precautions or with specific warnings elsewhere in this manual violates safetystandards of design, manufacture, and intended use of the instrument. Page 51 Specifications – ATable A ComiXology Thousands of Digital Comics.